Otherwise, if speed is keithley labview important you can do everything with LabVIEW controlling the timing of measurement using dataflow. I originally thought that I was ultimately limited by the fact that i was using the read command instead of the internal buffer, but then I tried keithlwy program that you supplied in the previous thread graphdatafast. You could simplify your life with a pair of Keithley s, s or s. I haven’t wired the inputs to the VISA Configure, because you’ve shown the defaults in your image – I’m sure you already checked but these should match the settings you have on the device – it doesn’t seem to list defaults in the manual. I am thinking the is taking too long to make the measurement. Message 10 of
|Date Added:||17 May 2010|
|File Size:||38.52 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I will try the steps your suggested. Usually, you can do this labviww Concatenate Strings, or similar. One example is zero correction see the second screenshot of the Kabview Tree below.
Solved: Labivew program for KE picoammeter – Discussion Forums – National Instruments
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. I have not been able to find anything that provides much detail. Otherwise, if speed is keithley labview important you lagview do everything with LabVIEW controlling the timing of measurement using dataflow.
Do you know the parameters the device has configured?
Instrument Control (GPIB, Serial, VISA, IVI)
Message 6 of If you have that kind of adaptor, you can use a null-modem adaptor to switch them back or use a different usb adaptor I will post code.
Please clarify what you mean by the block diagrams do not show up. Google [Bot] and 0 guests. Just a picoampmeter is not sufficient for measuring cv.
KEITHLEY 6485 LABVIEW DRIVER DOWNLOAD
It just uses keithlej loop with a READ? Could you please help me to correct the program if something is wrong?
I also tried to delete everything from keithley labview loop except the read function and an indicator of the measurement, turned auto rage off and set it to 0. You have to setup a hardware triggering between the Keithley units. Hi Please describe which other instrument you are using.
Keithley did not provide the cable. Message 4 of I have 3 terminal device like phototransistor. Assuming that you’re labbview past all of that, then you should have something you can communicate with nicely.
Keithley Picoammeter – Instrument Driver – National Instruments
So start small and learn how to go on by growing. Taking a keihtley look at the manual for the KeithleyI read these sections: It does not work.
Message 10 of For example, a simple picoamp Read test: You just increment the voltage and measure current. Message 2 of